Personal Health Device Implementation Guide
1.0.0 - STU 1

Publication Build: This will be filled in by the publication tooling

## Generating the PHD codes

These codes are used to express the Continua Personal Area Network (PHD) interfaces. The codes express both the transport (Bluetooth, USB, etc.) and specialization (Blood pressure, thermometer, glucose, etc.).

The codes are generated by the following relation:

PHD transport code = transport code X 8192 + (specialization term code - 4096)

The transport code (Tcode) has the following values:

Transport Code | Transport
--- | ---
0 | Continua version 1.0
1 | USB
2 | Bluetooth HDP
3 | ZigBee
4 | Bluetooth Low Energy
5 | NFC

The special Tcode of 0 is for Continua version 1.0 when there was no transport component in the reported certified PHD interface codes.

A sample of the most common specialization term codes

Specialization | MDC term code | Reference Identifier
--- | --- | ---
Generic 20601 | 4169 | MDC_DEV_SPEC_PROFILE_GENERIC
Pulse Oximeter | 4100 | MDC_DEV_SPEC_PROFILE_PULS_OXIM
Electro cardiograph | 4102 | MDC_DEV_SPEC_PROFILE_MIN_ECG
Blood Pressure Cuff | 4103 | MDC_DEV_SPEC_PROFILE_BP
Thermometer | 4104 | MDC_DEV_SPEC_PROFILE_TEMP
Respiration rate | 4109 | MDC_DEV_SPEC_PROFILE_RESP_RATE
Weight Scale | 4111 | MDC_DEV_SPEC_PROFILE_SCALE
Glucose Monitor | 4113 | MDC_DEV_SPEC_PROFILE_GLUCOSE
Coagulation meter | 4114 | MDC_DEV_SPEC_PROFILE_COAG
Insulin Pump | 4115 | MDC_DEV_SPEC_PROFILE_INSULIN_PUMP
Body Composition Analyizer | 4116 | MDC_DEV_SPEC_PROFILE_BCA
Peak Flow meter | 4117 | MDC_DEV_SPEC_PROFILE_PEAK_FLOW
Sleep Apnea Breathing Equipment | 4120 | MDC_DEV_SPEC_PROFILE_SABTE
Continuous Glucose Monitor | 4121 | MDC_DEV_SPEC_PROFILE_CGM
Cardiovascular Device | 4137 | MDC_DEV_SPEC_PROFILE_HF_CARDIO
Strength Equipment | 4138 | MDC_DEV_SPEC_PROFILE_HF_STRENGTH
Independent Activity/Living Hub | 4167 | MDC_DEV_SPEC_PROFILE_AI_ACTIVITY_HUB
Medication Monitor | 4168 | MDC_DEV_SPEC_PROFILE_AI_MED_MINDER